NI-VISA USB DEVICE DRIVER DETAILS:
|File Size:||9.6 MB|
|Supported systems:||Windows XP (32/64-bit), Windows Vista, Windows 7, Windows 8.1, Windows 10|
|Price:||Free* (*Free Registration Required)|
NI-VISA USB DEVICE DRIVER
When a matching. USB 3.
Identifying Multiple Identical USB Raw Devices with NI-VISA
USB Instrument Control Tutorial - National Instruments
USB communicates using four types of pipes or endpoints: control, bulk, interrupt, and ni-visa usb device. Each type of pipe transfers a different type of information. Furthermore, any number of endpoints can be of any endpoint type. Think of an endpoint as a communication socket. In Windows this binding is done using.
Search for your device to see if there is a download available. Yes No Submit. I have a device with USB port.
VISA's greatest advantage is that it is an extremely easy language to learn. A window similar to that shown in Figure 10 ni-visa usb device open. Double-click on your device to open a VISA session to the device.
The window shown in Figure 11 should open. To read the properties of your device, select the View Attributes tab.
This lists the different attributes of ni-visa usb device device, including information like Resource Name. Both are available through the links at the end of this tutorial.This tutorial is meant as a starting point for using NI-VISA to communicate with a USB device. It is not intended as a starting point for learning.
I noticed that NI-VISA introduced USB support. How can I get started communicating with a USB device using the RAW mode?.